If everything is correct the power supply will now be changed to 5. Open these examples using the following procedure. Can I do something like that for the board? Also, I’ve heard of tools that allow you to “draw” the waveform and save it in an image file and then feed that image file into the function generator. After above, I opened “ag33xxx – Getting Started 2. Once you get this part to work, look up some of the other commands in the manual on page There is also an RS connection serial port but that is not of use for the equipment that is at hand.
|Date Added:||2 December 2009|
|File Size:||10.49 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
The instrument is now configured. Also, I’ve heard of tools that allow you to “draw” the waveform and save it in an image file and then feed that image file into the function generator. Notice that as the mouse is moved over each of the different blocks of the windows, text at the top of the window shows what the block contains. Here the validation test bed uses an Agilent Logic analyzer setup which is currently operated manually.
Hundreds of built-in functions are included with LabVIEW and the Agilent 33XXX series driver After placing the Configure Burst labvoew, be sure to connect the purple instrument wire and the yellow error wire appropriately. If you use some labvieew programming you can get the program to respond just by changing a number here, or the push of a lwbview there. The return message starts with?
Also notice then when you get into a submenu of the main menus, the blue arrow on the top left of the window becomes active. Try it out sometime, because it will come in handy later. This may take some trial and error, maybe even some reading on your part to get this to work. Have your TA initial your sheet.
laview Firstly, that’s my mistake to not change scope setting “Probe scale” to 0. This also comes in handy when a variety of measurements are needed over a large range of values. The Front Panel and the Diagram are shown in Figures 1 and 2 respectively. Hello,I used labview to call a dll, but it have one parameter “pbyFwData” which Pointer to the firmware binary data stream.
I have written the code but its showing Error Can I do something like that for the board?
Customize the panel of the VI to fit your application needs. This way the experimenters can concentrate on the data they are collecting instead of worrying about how everything is set up and changing dials and knobs on all the equipment. Use the following procedure to add manual inputs for noise, samples, and frequency: If the frame displayed is not Frame 0, change it to Frame 0.
This tool automatically detects USB hardware connected to your host PC and adds a graphical representation to the window. What if there was a way to set up a computer program to change and measure all of the different measurements? Le probleme c’est que la fruence d’hantillonage de ma boucle while a beaucoup chut et mon programme ne fonctionne plus correctement.
I have successfully used several Generators Would you please send the code to this forum – then we can discuss about it. Rate this document Select a Rating 1 – Poor 2 3 4 5 – Excellent.
Agilent Technologies / Keysight Technologies 33250A
You can also troubleshoot and make manual function calls to your instrument to ensure that it is working as expected before moving into automation. Import ascii file into Simulate Arb. To make a sinusoidal waveform use 332250a command Sin 5. Then I opened LV ag33xxx.
LabVIEW driver for the A | Keysight (formerly Agilent’s Electronic Measurement)
Has anyone found a way to do this? The next step is optional and is not required, but could help in finding errors that occur in the communication with the device. Open these examples using the following procedure.
This will obviously cause albview simulation to run until it is stopped by the user. When I try to mass compile the Test Exec llb’s, I get many errors indicating that some of the vi’s have no block diagram, so they cannot be upgraded.